The Washer Disinfector is a heavy-duty, high volume, single chamber, cabinet style washer designed for thorough and efficient cleaning of laboratory glassware, utensils, and miscellaneous items used in research and healthcare facilities.
LYNX 610LX Surgical Instrument Washer Disinfector size
Surgical Instrument Washer Compartment usable space: 26” wide x 26" high x 29" deep
Surgical Instrument Washer Overall Unit*”: 42" wide x 80" high x 36" deep Sliding Door(s)
LYNX 610LX Surgical Instrument Washer load and unload configuration: (select one):
Washer Disinfector front load and unload: pass through load and unload The Washer Disinfector is provided with an unload door and an additional cycle status indicator on the unload end, allowing one-way product flow from the contaminated area to the clean area. Washer Disinfector shall be provided with a combination of rotating jet spray and direct spindle spraying for all treatment solutions. The Washer Disinfector spray system consists of a single rotating spray arm mounted above, and interchangeable spindle wash or a rotating jet spray arms mounted below for internal treatment of the racked items.
LYNX 610LX Surgical Instrument Washer safety features
Opening the Washer Disinfector load-end door (or unload-end if pass-through unit), automatically stops operation of the washer. The Washer Disinfector door(s) must be closed, alarm acknowledged, and the Cycle Start button depressed to continue operation. A Washer Disinfector Emergency Stop button is provided on the load-end control panel (and unload-end if pass-through unit). If the Washer Disinfector operations of the wash cycle are interrupted by depressing the Button, the operator must reset the button and push the “Cycle Start” button to continue operations from the beginning of the selected cycle.
LYNX 610LX Surgical Instrument Washer computer control panel
A Washer Disinfector solid-state microcomputer control system monitors and automatically controls all process operations and functions. A Washer Disinfector twelve-cycle treatment menu may be programmed and retained to permit operating personnel to accommodate a wide variety of loads and processing requirements. Individual cycle phase times and temperatures and other key process parameters are programmable. Each Washer Disinfector cycle may be locked by supervision to ensure process integrity. The Washer Disinfector Cycle programming may be controlled by a supervisory access code. Each cycle program may be reviewed on demand.
LYNX 610LX Surgical Instrument Washer automatic treatment cycles
The standard Washer Disinfector treatment cycle consists of any or all of the following phases: Pre-wash, Wash, Rinse, Wash II, Rinse and Final Rinse. All cycle phases may be selected or deselected by the user. All Washer Disinfector Wash and Rinses treatments are re-circulated under pump pressure except the optional non-re-circulated rinse. The cycle, once activated, is completely automatic.
LYNX 610LX Surgical Instrument Washer debris screens
Located in the Washer Disinfector sump, easily removable screens are provided to protect items and large debris from entering the recirculation system. Washer Disinfector WATER SAVINGS Water from the last rinse cycle may be held and utilized as the pre-wash of the next cycle.
LYNX 610LX Surgical Instrument Washer Disinfector times and temperatures
Surgical Instrument Washer Disinfector temperature guarantee:
The Washer Disinfector operator may select to guarantee the temperatures for any re-circulated phase of the cycle. If the Washer Disinfector temperature guarantee is selected, the phase will not begin timing until the re-circulated water temperature reaches the set point. If the Washer Disinfector re-circulated temperature drops below the set point during the phase, the timer will stop. Once the Washer Disinfector set point temperature is again met, the phase-timer continues from where it stopped thus assuring the set temperature during entire treatment phase time.
LYNX 610LX Surgical Instrument Washer Disinfector operation
The Washer Disinfector operator loads the items to be cleaned in the wash chamber, closes the door, selects the proper cycle, and depresses the Cycle Start button. The Washer Disinfector proceeds through the treatment schedule and automatically shuts off at the completion of the cycle. The Washer Disinfector operator then opens door and removes the cleaned items.
LYNX 610LX Surgical Instrument Washer Disinfector treatment schedule
Surgical Instrument Washer PRE-WASH
Either Hot water from the house supply fills the Washer Disinfector sump, or Water remaining in the chamber sump from final rinse of the previous cycle is re-circulated through the jet system under pump pressure and pumped to drain upon completion. Phase is adjustable from 0 – 9999 seconds in 1-second intervals and 120F - 190F in 1F intervals.
LYNX 610LX Surgical Instrument Washer Disinfector detergent wash
Hot water from the house supply fills the Washer Disinfector sump and is pumped through the jet system. As required, a specified amount of detergent is added during recirculation using the OPTIONAL LYNX- supplied, or customer supplied detergent injection pumps. At the end of the treatment, the detergent solution is pumped to drain. Phase is adjustable from 0 to 9999 seconds in 1-second intervals and 120F - 190F in 1F intervals.
LYNX 610LX Surgical Instrument Washer Disinfector rinse
Hot water from house supply fills the Washer Disinfector chamber sump and is re-circulated through the jet system under pump pressure to remove residual detergent. Re-circulated water is pumped to drain on completion of treatment. Hot water from house supply fills the Washer Disinfector chamber sump and is re-circulated through the jet system under pump pressure to remove residual detergent. Re-circulated water is pumped to drain on completion of the Washer Disinfector treatment.
Surgical Instrument Washer FINAL SANITARY WATER RINSE
The Washer Disinfector Load is sprayed with either re-circulated or an OPTIONAL non-re-circulated sanitary water system. At the end of the Washer Disinfector treatment, the water may be retained in the chamber sump for use as the pre-wash water in the subsequent load or directed down the facility drain. The Washer Disinfector Re-circulated phase is adjustable from 0 – 9999 seconds in 1-second intervals and 120F - 190F in 1F intervals.
LYNX 610LX Surgical Instrument Washer Surgical Instrument Washer construction
The Washer Disinfector base, wash chamber and chamber sump are constructed of 304 Stainless Steel. The Washer Disinfector frame and cabinet, door and hardware are # 304 Stainless Steel. The Washer Disinfector is provided with vertical sliding doors.
LYNX 610LX Washer Disinfector Surgical Instrument Washer features
LYNX 610LX Washer Disinfector Surgical Instrument Washer optional features
The Washer Disinfector is equipped with a dryer. Heated air is re-circulated throughout the chamber at a user-selected temperature: The Washer Disinfector is provided with automatic power activated doors. The Washer Disinfector operator may open and close the door(s) by the push of a button. If the Washer Disinfector is a double door (pass-through) configuration the automatic door logic will be interlocked to allow only one door to be opened at a time. The Washer Disinfector is provided with a cold-water inlet valve, integral with the drain line to add cold water during draining for assistance in reducing the effluent temperature.
Surgical Instrument Washer Disinfector Surgical Instrument Washer accessories
The Washer Disinfector is provided with a stainless steel trim flange to enclose the opening between one end of the washer and the masonry wall opening. A stainless steel modular wall assembly is designed and manufactured to enclose an open wall area or to create a wall barrier. All Washer Disinfector materials and fasteners are of stainless steel construction. A stainless steel transfer cart is provided to connect to the load or unload end of washer for accessories transportation and ease of loading of the washer.